The IFC signal converter is designed to measure the flow velocity, IFC signal converter can be combined with any KROHNE flow. The Krohne IFC signal converter will need to be setup to output pulses from the flow meter. This guide will outline the procedure on the IFC signal. Ifc • Read online or download PDF • KROHNE IFC Converter Quickstart EN User Manual.
|Published (Last):||13 July 2007|
|PDF File Size:||1.77 Mb|
|ePub File Size:||18.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
Open the catalog to page 3. Current not correct, mA output cable has. This email is already in use.
Operation – KROHNE IFC 100 Converter EN User Manual
If you require data that is more relevant to your specific application, please contact us or your local sales office. Forgot user name or password? You have one of those, right? F flow exceeding limit. F over range A The current or the corresponding measured value is limited by a filter setting.
Measuring pipe not filled, function dependent on Fct. That means it should have a minimum inlet and outlet run, a pipe filled with product, and an upward orientation. Would you like to sign in? Then when you switch to your desktop, you get deluged with too much information, taking way too long to find anything. The current at the current input is less than. Could you give me the right link to download?. F backup 2 settings. Upload backup 1 or backup 2 settings, check. No middle ground here!
Open the catalog to page 6. F factory settings Error during the CRC check of the factory settings. Wir freuen uns, dass Sie ein Mitglied geworden sind!
Application-dependent fault, but device is. The signal converter can be combined with any measuring sensor, making it very widely used.
Error during the CRC check of the backup 1.
Application fault, device OK, but measured values affected. The difference lies in whether the measured value is also set to zero upon manyal detection of an empty pipe.
Error during the Ifcc check of the active. F application error Application-dependent fault, but device is OK. Diagnostic function, conductivity measurement and simple navigation to name but a few. Send us your email, and we’ll send you the further instructions. Complain wrong Brand wrong Model non readable.
F backup 1 settings.
KROHNE IFC Converter Quickstart EN User Manual | 32 pages
For remote signal converters, check the connection of the signal cable. F empty pipe 1 or 2 measuring electrodes are not in contact with the medium; measured value is set to zero.
Application test or operator action necessary. But before I drop that link, let me introduce the class to your converter.
Field frequency is not reaching steady state. It is a good idea to protect the converter from external heat sources such as direct sunlight as higher temperatures reduce the life cycle of all electronic components. If you really need the link manul now, then just scroll to the next header. Now create your krohe. No message if empty pipe.
Group message, when errors as described. Inside of the fluid, a voltage U is generated: Open the catalog to page 4. Open the catalog to page 1. Current not correct, mA output cable has open circuit or load too high.